DNA/RNA Extraction Kit (Performance)
HHMag Viral DNA/RNA Extraction Kit
Packing Specification 16T/plate×2 plates,16T/plate×4 plates、96T/box.
For nucleic acid extraction, enrichment, purification, and other steps. The treated products are used for clinical in vitro detection.
This product utilizes the specific adsorption capacity of magnetic beads for nucleic acid in the environment of high ionization salt to enrich nucleic acid in the sample lysates. After washing for removing impurities, the enriched nucleic acid is released in the eluate to achieve the purpose of nucleic acid purification.
1. Each functional component of HMD3602 and HMD3612 has been packed in deep hole plate in preloading, Lysis buffer 450 μL/test, Wash buffer 1 500μL, and Wash buffer 2 700 μL×2 test, Magnetic beads 200 μL/test, and Eluate 50 μL/test.
2. HMD3502 put the corresponding amount of Lysis buffer (450 μL/test), Wash buffer 1 (500 μL/test), and Wash buffer 2 (1400 μL/test). Eluent (50 μL/test), Magnetic beads (200 μL/test) in different containers, and separate them into the deep well plate or other containers by the operator before use.
3. Reagent Specifications:
Storage Condition and Validation Date
Keep storage at 2℃-25℃. Freezing can greatly reduce the performance of magnetic beads. Please avoid freezing during storage and transportation. This reagent is valid for 12 months under appropriate storage. Use the pre-loaded reagent as soon as possible after opening it.
Allsheng Auto-Pure 32A automatic nucleic acid extractor and similar instruments.
1. This product is suitable for nucleic acid extraction of cells and pathogens from nasopharyngeal swabs, saliva, urine, plasma, serum, cell preservation fluid, and other liquid samples.
2. Immediately after collection, samples should be placed into a cell preservation solution that can be used for nucleic acid detection for fixation and preservation, and nucleic acid extraction should be carried out as soon as possible.
3. Nuclease can affect the performance of this product
4. Saliva sample: take more than 200μL saliva sample, if less than 200μL, please use all.
5. Dry swab sample: add 400μL PBS (sample storage solution with sample swab) to sample tube with swab head, vortex for 2min, then use 200μL supernatant after centrifugal at top speed.
6. Wet swab sample: mix with high-speed vortex for 1min, absorb 200μL for use (if the liquid is less than 200μL, add the appropriate amount of PBS or sample storage solution with the sampling swab, mix with vortex and centrifuge)
7. Other liquid samples: more than 200μL take 200μL, if less than 200μL, please use all.
1. Automatic nucleic acid extraction instruction (For automatic extraction methods, this description takes HMD3602 and Allsheng Auto-Pure32A automatic nucleic acid extraction instrument as an example.
2. 96-well deep hole plate and magnetic rod sleeve (HMD3502 automatic extraction). 3. Magnetic rack (HMD3502 manual extraction).
1. Please take out the buffer from the refrigerator, and use it till room temperature.
2. If there is a small amount of precipitation in the lysis buffer, please place it at room temperature or 37℃ and use it after the precipitation is fully dissolved.
3. Before use, magnetic beads must be shaken until there is no visible precipitation at the bottom before being used.
HMD3602/HMD3612 automatic extraction
1. Unseal: take out the deep-hole plate from the packaging box, balance to room temperature, carefully tear off the sealing film, if there is liquid adhesion to the deep-hole plate sealing film and tube wall, throw the liquid to the bottom and then unseal.
2. Add sample: Add 200 μL sample to column 1/7 (HMD3602) or plate 1 (HMD3612), one sample for each well position. Note: please add sample within 20 minutes after opening.
3. Load in the machine: set extraction procedures according to the supplemental table. Put the deep-hole plate into the nucleic acid extraction instrument, insert the magnetic rod sleeve into the corresponding slot, and start the program. After the procedure was completed, the eluent was transferred to a centrifuge tube and the product was stored at -80℃. Discard waste according to standard procedures.
HMD3502 manual extraction
1. Add 200 μL samples and 450 μL lysis buffer into a 1.5 mL centrifugation tube, mix them upside down, and incubate at 55℃ for 5 minutes.
2. Add 200µL magnetic beads, rotate, and incubate at room temperature on a rotating blender for 10 minutes.
3. Place the mixture containing the sample on the magnetic rack and let stand for 2 minutes (or until the lysis buffer is colorless and transparent). Suck out the liquid phase and discard. Note: If there are magnetic beads on the cover of the centrifugation tube, the magnetic beads can be washed off by reversing the centrifugation tube with the magnetic rack 3-4 times during the standing process.
4. Remove the magnetic rack and add 500 µL wash buffer 1. Mix upside down until the beads are fully suspended or blow the beads 3-4 times with a pipette.
5. Place the centrifugal tube on the magnetic rack, stand for 1 minute and discard the supernatant.
6. Remove the magnetic rack, add 700 µL wash buffer 2, and re-suspend the magnetic beads.
7. Place the centrifugation tube on the magnetic rack, stand for 1 minute and discard the supernatant.
8. Repeat Steps 6 and 7. Note: Try to ensure that the residual liquid in the centrifugation tube is completely removed (including the residual liquid on the cover of the centrifugation tube).
9. Keep the centrifugation tube on the magnetic rack, open the cover and let stand at room temperature to dry for 5 minutes.
10. Remove the magnetic rack, add 50µL eluent, and re-suspend the magnetic beads. Incubate at 55℃ for 5 min, and gently flick the centrifugation tube 2-3 times during the incubation period to separate the magnetic beads Scattered.
11. Place the centrifugation tube on the magnetic rack and let stand for 2 minutes. Transfer the supernatant containing elution nucleic acid into the new centrifugation tube. Note: If there is liquid attached to the wall of the centrifugation tube, please immediately centrifugation before magnetic suction operation.
12. Discard the experimental waste liquid as required, and store the purified nucleic acid samples inappropriate conditions or conduct the next experiment.
HMD3502 automatic extraction